A new version of BeamformX has been posted on http://www.optinav.info/ftp.html .
This version is slightly different from 5.100, posted yesterday, but more detailed information that applies to both is given here:
For some users, the most important change is that some equalization has been applied to attempt to compensate for frequency response of the microphones in the ACAM 100 and ACAM 120. I based the corrections on the published datasheets for the two microphone types. Significant changes occur below 200 Hz and above 5 kHz. Some extrapolation of the published curves was required.
The array shading and the overload light thresholds have changed slightly.
When there is an ROI in the display, the near-real-time narrowband spectrum shows the result of beamforming to the point in the center of the ROI. The method for this beamforming has changed from delay and sum beamforming to conventional, frequency domain, beamforming (FDBF). This gives much better beamformed spectra. Note that the limitations of FDBF apply: poor dynamic range and low frequency resolution and treatment of coherent sources. Better spectra can be obtained by doing a frequency sweep with Quantitative Beamforming, but not yet in near real time. The old delay and sum calculation is still used for the audio output.
The near-real-time spectra creates the opportunity to compute a near-real-time dBA value for the center of the ROI. If Settings/A-weighting is selected, then this is shown in the upper right corner of the Display as the "20 Hz - 20 kHz" value. A number was shown there starting from Version 5.003, but it always gave the median of the microphones. Now it can be focused by creating the ROI.
The old "20 Hz - 20 kHz" dBA was found to be badly wrong in some cases. Several contributors this problem have been identified and corrected, including the microphone frequency response and some software bugs. Even with the corrections, the ACAM/BeamformX system is likely to give different results from those of a quality sound level meter. Some of the reasons are good.
A version of BeamformX called DroneHound is being developed to identify drones. A new version has been posted, but it is not ready for real use.
Known bugs/requests that are not yet fixed:
Sometimes clicking on the spectrum does not adjust the tuning frequency. This happens so rarely that it is difficult to troubleshoot.
The Spectrogram shrinks in between runs.
There is no scripting language or way to load new data without exiting BeamformX.
The language strings need updating.
There is no way to reopen the Spectrogram or Spectrum windows after closing them.
Higher resolution optical images are possible.
BeamformX does not look like a Windows program.
In this category, the material is of hard type. The material is categorized as unable to bend and reshaped. This material is inflexible. That’s why the rigid packaging can maintain the structure of the packed thing and give support. It is mostly known for its strength and durability. In the category of rigid, we will discuss the following packaging style:
Magnetic closure Boxes
Lid off Boxes
Version 5.101 has been replaced with 5.101a because a packaging error kept 5.101 from starting. The new sendspace link is https://www.sendspace.com/pro/dl/c0pen1